Lubrication \ Installation and Debugging
1. Characteristics of Chinese medicine ultra-fine crusher 1. The machine's motion system is continuously rotated, from the motor → belt drive → activity The disc is high-speed rotational motion, and the transmission is smooth when the high speed is rotated.
2. The front cover of the Chinese medicine ultrafine crusher must be opened after the stop, and the cavity is cleaned to ensure the cleaning of the machine work surface.
3. The broken material can be discharged directly in the abdomen of the Chinese medicine ultrafine pulverizer.
4, the thickness of the crushed degree can be achieved by a "Aperture) screening (Tōng Guò). Airflow pulverizer and cyclone separator, dust collector, and airplane set into a set of pulverization systems. After the compressed air is dried, it is sprayed into the pulverizing chamber by the Lavar nozzle, and the material is repeatedly collided with the intersection of the multi-high pressure airflow.
The crushing material is pulverized after the fan pulling force Airflow movement to the grading area, under the power of high-speed rotation, the thick centrifugal force generated by the high-speed rotation, the fine particles that meet the particle size requirements are collected by the grading wheel into the cyclone and the dust collector, and the crude particles drops to the pulverizing region to continue pulverization. . Second, the lubricating spindle of the ultra-fine crusher of the Chinese medicine must be added regularly, ZG-2, only the front and rear bearings (a new oil is changed every 3 months).
Third, the installation and commissioning of the ultrafine pulverizer of the Chinese medicine 1, the ultra-microfin crusher must be stable after the traditional Chinese medicine ultrafine crusher is stable and secured by the ground screw. The small Soviet research mill can be used to become an inert gas protection apparatus for an inert gas protection device such as a nitrogen gas such as nitrogen, a carbon dioxide gas.
2, before the test, the inner cavity of the machine should be cleaned and lock the front cover with a star handle.
3. After the power is turned on, it should be noted that the proprigation is turned to rotate in the direction shown.
4, should pay attention to whether the machine has abnormal noise when testing in order to write timely.
The small Soviet research mill can be used to become an inert gas protection apparatus for an inert gas protection device such as a nitrogen gas such as nitrogen, a carbon dioxide gas. 5, at the crack, press the storage bag.